CS ITCK System

IEC6100046 CS system
• Compact: Compact system bundle with easy setup.
• Modular: Future extendable, upgradeable & easy service.
• Flexible: Can be adopted to customer specific CI test needs, for example additional power meters for reverse power and sensing power.
• Stateoftheart: Shared and proven technology from DARE


RadiField (EField Generater)

RadiField´Â RF Power Amp + Antenna + Power Meter¸¦ ³»ÀåÇÏ¿©
ÀüÅëÀûÀÎ ½Ã½ºÅÛ¿¡¼ ¹ß»ýÇÏ°í Cable loss / Connector lossµîÀ» ¾ø¾Ù
¼ö ÀÖÀ¸¸ç, ±âÁ¸ÀÇ ¾î¶² ¾ÈÅ×³ª º¸´Ù BeamWidthÀ» ³Ð°Ô ¸¸µé ¼ö ÀÖ¾î
IEC6100043¿¡ µû¸¥ ½ÃÇè¹üÀÇ ½±°Ô ¼öÇà ÇÒ ¼ö ÀÖÀ½.
> ÃÖ´ë 3m °Å¸®¿¡¼ 1GHz~6GHz uniformity 10V/m(18V/m),
> 1GHz~6GHz 1m °Å¸®¿¡¼ 54V/m)


DARE Instrument ¼Ò°³

EMC equipment and EMC software
DARE!! Instruments develops and produces high quality EMC equipment like EMC
measurement
instruments and full automated EMC test software. DARE!! Instruments has developed a number
of unique
products in the field of EMC measurements and calibrations.


RadiMation¢ç

* The integrated software for automated EMC measurements.
* The software enables
fast and accurate EMC measurements
and supports all commercially available EMC
equipment.


RadiCentre¢ç

*The one, two or sevenslot 19inch rack mountable system.
* For RadiSense¢ç,
RadiPower¢ç, RadiGen¢ç, RadiControl¢ç, RadiMate¢ç, RadiLink¢ç and RadiSwitch¢ç
modules.


RadiSense¢ç:

* The unique family of batteryfree (laser supplied) isotropic e field sensors
allows measurements
with field strengths of 0,25 V/m till 4000 V/m in the
frequency range of 10 kHz to 18 GHz.
* Its small dimensions make it the ideal
probes for measurements in small cells and nonstop measurements.


RadiGen¢ç

* Family of EMC / RF Signal Generators.


RadiLink¢ç

* The optical alternative for the coaxial cable with wide dynamic range up to 3
GHz.


RadiSwitch¢ç

* The solution for low loss switching of high frequency (RF) signals.
*Several
switch cards available with SPDT^and SP6T relays in BNC, Ntype, SMA type
(18GHz) and Ktype (40GHz) with power levels up to 700W.
* All cards are intended
for use in the RadiCentre¢ç.


RadiPower¢ç

* The successful range of EMC / RF power meters for EMC and Telecom applications.
*From 9 kHz to 18 GHz, CW, Burst/Pulse and Wireless versions.
Due to the high
speed, even at low power, it is the ideal power meter for fast and accurate EMC
measurements.


RadiMate¢ç

* The four channel 16 bit A / D and DAconverter, for use in anechoic rooms and
cells,
to monitor EUT signals during EMC measurements.
* Immune to 300 V / m.


RadiControl¢ç

* The universal controler card for EMC turntables and antenna masts


RadiTurn¢ç

* The low profile, transparent turntable solution for EMC measurements and
calibrations in anechoic rooms and Open Area Test Sites (OATS).


HI4413P Fiber to Serial Converter

Key Features
 Can also be used with HI3702, HI3638
 Provides Isolation from Field Strength Measurement Point
 Used with most Holaday optically coupled probes.
 No External Power Supply Required


2020 MultiAxis Positioner (MAPS)

Key Features
 Angular positioner accuracy within +/ 0.25¡Æ
 Variable speed rotation
 MultiAxis positioner
 Fiber optic control lines eliminate RF noise
 360¡Æ independent rotation in both azimuth and ortho axes


GN7000

* 100 kHz  200 MHz / 7000 Watts
* Class A solid state
* Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB >5000 W and H<20dBc up to 100MHZ P1dB >2000 W and H<20dBc from 100MHZ to 200MHZ


LT90

* 20 kHz  1000 MHz / 90 Watts
* Class A solid state
* Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB >60 W and H<20dBc P1dB >75 W and H<20dBc P1dB >45 W and H<20dBc


LT320

* 20 kHz  1000 MHz / 320 Watts
* Class A solid state
* Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB >280 W and H<20dBc from 20MHZ to 500 MHZ P1dB >200 W and H<20dBc from 500MHZ to 1000


LT600

* 20 kHz  1000 MHz / 600 Watts
* Class A solid state
* Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB >350 W and H<20dBc from 30MHZ to 80 MHZ
P1dB >600 W and H<20dBc from 80MHZ to 600 P1dB >350 W and H<20dBc from 600MHZ to 1000


MT100

* 80 MHz  1000 MHz
*Class A solid state
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 80 W and H < 20 dBc up to 500 MHz and
P1dB > 60 W and H < 20 dBc from 500 MHz to 1000 MHz


MT200

*80 MHz  1000 MHz / 200 W CW
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 140 W and H < 20 dBc up to 600 MHz
P1dB > 110 W and H < 20 dBc from 600 MHz to 1000 MHz


MT400

*80 MHz  1000 MHz
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 350 W and H < 20 dBc up to 500 MHz
P1dB > 220 W and H < 20 dBc from 500 MHz to 1000 MHz


MT700

*80 MHz  1000 MHz
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 600 W and H < 20 dBc up to 500 MHz
P1dB > 400 W and H < 20 dBc from 500 MHz to 1000 MHz


MT1200

*80 MHz  1000 MHz
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 1000 W and H < 20 dBc up to 500 MHz
P1dB > 700 W and H < 20 dBc from 500 MHz to 1000 MHz


MT1400

*80 MHz  1000 MHz
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 1200 W and H < 20 dBc up to 400 MHz and
P1dB > 700 W and H < 20 dBc from 400 MHz to 1000 MHz


MT1700

*80 MHz  1000 MHz
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 1500 W and H < 20 dBc up to 600 MHz
P1dB > 1200 W and H < 20 dBc from 600 MHz to 1000 MHz


MT2000

*80 MHz  1000 MHz
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 2000 W and H < 20 dBc up to 500 MHz
P1dB > 1200 W and H < 20 dBc from 500 MHz to 1000


MT2800

*80 MHz  1000 MHz
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 2700 W and H < 20 dBc up to 500 MHz
P1dB > 1600 W and H < 20 dBc from 500 MHz to 1000 MHz


SV20

*800 MHz  3200 MHz
*Class A solid state
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 14 W and H < 20 dBc


SV40

*800 MHz  3200 MHz/40 W CW
*Class A solid state
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 30 W and H < 20 dBc up to 2 GHz
P1dB > 25 W and H < 20 dBc from 2 GHz to 3.2 GHz


SV70

*800 MHz  3200 MHz /70W CW
*Class A solid state
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 70 W and H < 20 dBc up to 2 GHz
P1dB > 40 W and H < 20 dBc from 2 GHz to 3.2 GHz


SV120

*800 MHz  3200 MHz
*Class A solid state
*Typical output power : 120 W CW
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 90 W and H < 20 dBc up to 2 GHz
P1dB > 70 W and H < 20 dBc from 2 GHz to 3.2 GHz


SV220

* 800 MHz  3200 MHz
*Class A solid state
*Typical output power : 220 W CW
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 180 W and H < 20 dBc up to 2 GHz
P1dB > 160 W and H < 20 dBc from 2 GHz to 3.2 GHz


SV450

*800 MHz  3200 MHz
*Class A solid state
*Typical output power : 450 W CW
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 300 W and H < 20 dBc up to 2 GHz
P1dB > 250 W and H < 20 dBc from 2 GHz to 3.2 GHz


SV1000

* 800 MHz  3200 MHz
* Class A solid state
* Typical output power : 1000 W CW
* Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB >570 W and H<20dBc up to 2 GHZ
P1dB >470 W and H<20dBc from 2 GHZ


SW18

*800 MHz  4000 MHz /18 W CW
*Class A solid state
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 13 W and H < 20 dBc up to 2.5 GHz
P1dB > 12 W and H < 20 dBc from 2.5GHz to 4 GHz


SW35

*800 MHz  4000 MHz / 35W CW
*Class A solid state
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 25W and H < 20 dBc up to 3GHz
P1dB > 20 W and H < 20 dBc from 3GHz to 4 GHZ


SW65

*800 MHz  4000 MHz /65 W CW
*Class A solid state
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 40W and H < 20 dBc up to 3GHz
P1dB > 30 W and H < 20 dBc from 3GHz to 4 GHZ


SW100

*800 MHz  4000 MHz / 100 W CW
*Class A solid state
*Broadband (instantaneous single band): 0.8 GHz – 4 GHz
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
p1dB > 70W and H < 20 dBc up to 3GHz
P1dB > 60 W and H < 20 dBc from 3GHz to 4 GHZ


SW200

*800 MHz  4000 MHz / 200W CW
*Class A solid state
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 140W and H < 20 dBc up to 3GHz
P1dB > 120 W and H < 20 dBc from 3GHz to 4 GHZ


SW420

*800 MHz  4000 MHz /420 WCW
*Class A solid state
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 300W and H < 20 dBc up to 3GHz
P1dB > 220 W and H < 20 dBc from 3GHz to 4 GHZ


SW800

*800 MHz  4000 MHz / 800 W CW
*Class A solid state
*Typical output power : 800 W CW
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 520W and H < 20 dBc up to 3GHz
P1dB > 420 W and H < 20 dBc from 3GHz to 4 GHZ


UX15

*2000 MHz  6000 MHz
*Class A solid state
*Typical output power : 15 W CW
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 12W and H < 20 dBc up to 4GHz
P1dB > 11W and H < 20 dBc from 4GHz to 6GHZ


UX30

*2000 MHz  6000 MHz
*Class A solid state
*Typical output power : 30 W CW
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 20W and H < 20 dBc up to 4GHz
P1dB > 17W and H < 20 dBc from 4GHz to 6GHZ


UX55

*2000 MHz  6000 MHz
*Class A solid state
*Typical output power : 55 W CW
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 40W and H < 20 dBc up to 4GHz
P1dB > 35W and H < 20 dBc from 4GHz to 6GHZ


UX100

*2000 MHz  6000 MHz
*Class A solid state
*Typical output power : 100W CW
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 70W and H < 20 dBc up to 4GHz
P1dB > 55W and H < 20 dBc from 4GHz to 6GHZ


UX150

*2000 MHz  6000 MHz
*Class A solid state
*Typical output power : 150W CW
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 100W and H < 20 dBc up to 4GHz
P1dB > 80W and H < 20 dBc from 4GHz to 6GHZ


SX40/15

*800 MHz  6000 MHz
*Class A solid state
*Typical output power : 40 W CW (0.8 – 3.2 GHz) and15 W CW (3.2 – 6 GHz)
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 25 W and H < 20 dBc up to 3.2 GHz
P1dB > 10 W and H < 20 dBc from 3.2 GHz to 6 GHz


SX40/30

*800 MHz  6000 MHz
*Class A solid state
*Typical output power : 40 W CW (0.8 – 3.2 GHz) and 30 W CW (3.2 – 6 GHz)
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 25 W and H < 20 dBc up to 3.2 GHz
P1dB > 16 W and H < 20 dBc from 3.2 GHz to 6 GHz


SX70/15

*800 MHz  6000 MHz
*Class A solid state
*Typical output power : 70 W CW (0.8 – 3.2 GHz) and 15 W CW (3.2 – 6 GHz)
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 40 W and H < 20 dBc up to 3.2 GHz
P1dB > 10 W and H < 20 dBc from 3.2 GHz to 6 GHz


SX70/55

*800 MHz  6000 MHz
*Class A solid state
*Typical output power : 70 W CW (0.8 – 3.2 GHz) and 55 W CW (3.2 – 6 GHz)
*Linear output power (1 dB compression) guaranted with harmonics <20 dBc:
P1dB > 38 W and H < 20 dBc up to 3.2 GHz
P1dB > 32 W and H < 20 dBc from 3.2 GHz to 6 GHz


Pince IPDR250

*Broadband (instanteous single band) : 10KHZ400MHZ
*Typical power: 500 W cW


Jig CJDR250

* Calibration jig for ipdr250


Data Sheet 352

Frequency Range: 10 kHz  5 GHz


Data Sheet 354

Frequency Range: 10 kHz  3 GHz


Data Sheet 317

Frequency Range: 10 kHz  2.5 GHz


Data Sheet 330

Frequency Range: 10 kHz  2.5 GHz


Data Sheet 310

Frequency Range: 9 kHz  1 GHz


Data Sheet 315

Frequency Range: 10 kHz  1 GHz
Low noise, very high gain

